Dear Fluke Customer:
Fluke is voluntarily recalling its 190 Series ScopeMeter® 100MHz and 200MHz 4 Channel Hand Held Oscilloscopes that contain firmware version V10.40. This includes all units manufactured between February 5, 2012 and April 3rd 2012 and have a serial number range of 1954xxxx to 2009xxxx, and all units upgraded to firmware version V10.40 after February 5, 2012.
Description of the problem
Four channel models with the latest firmware (V10.40) may display the decimal point of the measurement value in the wrong position. This poses a potential safety risk for the user, as an unsafe voltage may be indicated as a much lower (safe) voltage. THIS RISK COULD RESULT IN SERIOUS INJURY OR DEATH.
The affected models numbers are listed in the table below. You can determine whether your 4-channel ScopeMeter® is being recalled by checking the firmware version as described below. Stop using it until after that determination is made.
|Model Numbers||Description Ending Serial No.|
|FLUKE-190-104/xx||SCOPEMETER® 4 CHANNEL 100 MHZ COLOR|
|FLUKE-190-104/xx/S||SCOPEMETER® 4 CHANNEL 100 MHZ COLOR|
|FLUKE-190-204/xx||SCOPEMETER® 4 CHANNEL 200 MHZ COLOR|
|FLUKE-190-204/xx/S||SCOPEMETER® 4 CHANNEL 200 MHZ COLOR|
|Where xx is 2 letter code defining country or region specific line cord and user language set|
This recall does not include the 2 channel ScopeMeter® units.
Identifying the recalled product
To determine whether your ScopeMeter® is being recalled, first check the firmware version of the product by following these steps: (1) Disconnect the ScopeMeter® from any signal source; (2) with the unit turned on and facing you, press the user key located at the bottom right side of the instrument's front panel; (3) press F3 (Version and Cal).
The firmware version is then displayed as shown below with other instrument information
- Stop using the Fluke 4 channel ScopeMeter® even if you have not experienced problems.
- Verify your 4 channel ScopeMeter® is affected by this recall using the steps above.
- Either return your 4 channel ScopeMeter® for a free upgrade or go to Fluke 190 Fix and follow the instructions to download a new version of the firmware
You can return your ScopeMeter® by submitting an online service request or calling (888) 97FLUKE (35853), 7 a.m. to 4 p.m. US Pacific Time Monday–Friday. Follow the instructions provided and we will send you a prepaid mailer to return your unit at no cost.
If you are located outside the US, contact your local authorized service center.
We are undertaking this voluntary recall because we place tremendous importance on our customer's safety and the reputation of our products. Please accept our apologies for any inconvenience caused by this action.
James L. Cavoretto
Sr. Vice President
Chief Technical Officer