. . .
Testing insulation resistance is essential to keeping wires and motors working properly. Megohmmeters provide a quick and easy way to determine the condition of the insulation on wire, generators, and motor windings. A megohmmeter is an electric meter that measures very high resistance values by sending a high voltage signal into the object being tested. Typically, however, this is the only function a megohmmeter performs.
While megohmmeters are often informally referred to as insulation testers, strictly speaking this is not accurate. Why? What’s the difference between a megohmmeter and an insulation tester? An insulation tester performs the basic measuring function a megohmmeter does—measuring very high resistance values by sending a high voltage signal into the object being tested—and it often does much more; it usually performs more functions, including more complex testing and recording of measurements.
A full-featured insulation tester can perform high-voltage insulation resistance tests, and much more
For example, unlike megohmmeters, insulation testers can also measure voltage and current. The Fluke 1587 FC insulation multimeter, for example, can perform insulation tests up to 1000 volts, AND it is a full function digital multimeter. The Fluke 1550c can generate up to 5000 volts for insulation tests. Insulation testers can also perform more complex tests, such as compensating for ambient environmental conditions like humidity and temperature during a test to provide information about how motors perform in changing conditions. Because environmental conditions and/or chemical contamination accelerates the rate at which insulation degrades, it’s critical to compare insulation resistance test results that are corrected for different test conditions.
Insulation testers like the Fluke 1587 FC and the Fluke 1550c offer another advantage over megohmmeters. Memory storage through Fluke Connect® saves measurements to your phone or the cloud so you don’t have write down results. This saves time, reduces errors, and saves data for historical tracking over time.
Choosing between an insulation tester and a megohmmeter depends on your business needs. A meg test may be all you need. But if you want increased power, convenience, prevention, and safety, an insulation tester may be your best choice.
|Fluke 1587 FC Insulation multimeter||Fluke 1550c Insulation tester||Megger MIT230||Extech 380363|
|Test voltage||50 V, 100 V, 250 V, 500 V, 1000 V||250 V, 500 V, 1000 V, 2500 V, 5000 V||250 V, 500 V, 1000 V||250 V, 500 V and 1000 V|
|Resistance measurements||2.2 GΩ||2 TΩ||1 GΩ||10 GΩ|
|Data recording||Unlimited w/ Fluke Connect®||99 internal, unlimited w/ FC||Manual entry 9 records|
|Voltage measurement||0-1000V||25V – 600V||999V|
|Current measurement||400 mA ac or dc|
|Frequency measurement||99.99 kHz|
|Capacitance measurement||9999 μF||15 μF|
|Temperature measurement||-40 °C to 537 °C|
-40 °F to 998 °F
A deadweight tester is a proven method of pressure calibration that is usually chosen for bench applications when accuracy and reliability are the top . . .
Early reports from the field suggest that the new Fluke 772 and 773 Milliamp Process Clamp Meters may be game changers, breaking new ground for technicians . . .
Importance of investing in battery storage systems health and maintenance.
If your compressed air system is under-performing, your connected equipment will be too. Learn how to monitor and prioritize which air leaks to fix using . . .
Make energy efficiency a part of your preventive maintenance plan to make a positive impact on your bottom line, improve plant performance, and maintain . . .
At commercial buildings and industrial plants, energy costs can quickly add up, especially when older lighting systems are in place, motors age and . . .
Information on the role of thermal imagers in preventive maintenance programs, including cost savings, return on investment (ROI), and the inspection process. . . .
There are many ways to prioritize assets in a maintenance routine. Most approaches are incomplete or create more problems than they solve. Learn how to . . .
How to measure output voltage from a VFD to a motor
The arc flash boundary is the minimum “safe” distance from exposed energized conductors or circuit parts that has the potential for an arc flash. The . . .
Following electrical safety regulations in industrial plants is critical. But myths about electrical safety are rampant. Understand the facts vs myths . . .
If you troubleshoot or maintain equipment in an industrial setting, staying safe depends partly on the environment and partly on you.
This article demonstrates how to detect the 13 most common causes of winding insulation and bearing failure in advance.
Troubleshooting the most common power quality problems—voltage distortion and harmonics. Tools needed, types of problems
HVACR mechanics rarely rely on a single test to pass or fail a function or process. As they walk up to a condensing unit, they listen, they wave their . . .
Testing medical gases and piping systems in healthcare facilities and hospitals presents multiple challenges for maintenance teams. Learn why efficient . . .